Here is the procedure:
1. In Machine Setup add the rack or racks you are using.
2. Calibrate the rack(s)
3. Configure the Probe Station (rack) - For an FCR typr rack you can change the adapter types (empty for SM modules, PA25-SH for SP25 Stylus Holders, and PA25-20 for TP20's)
Now go to the Probe Assy Dialog to correlate each Probe Assy with a Rack Port. First add the SP25M an be sure to click Update Assembly.
4. Now add the SM25-1/SH25-1 or similar scanning module. You should see a dialog dislayed with an option to Assign to Port. Assign to Port 1
5. Now at the bottom of the Probe Assy dialog assign the entire Probe to Port 2.
Finally add another Probe Assembly and repeat steps 4 & 5. Be sure to assign the SM25-1 module to port 1 and the Probe Assy to Port 3.
This will allow changing just the Stylus Holder when changing from Probe Assy 0 to Probe Assy 1 while still using a common SM25 Scanning Module.